Cervical spine immobilisation can make direct laryngoscopy difficult, which might lead to airway complications. This randomised control trial compared the time to successful intubation using either the Macintosh laryngoscope or the McGrath? Series 5 videolaryngoscope in 128 patients who had cervical immobilisation applied. Intubation difficulty score, Cormack Lehane laryngoscopic view, intubation failures, changes in cardiovascular variables and the incidence of any complications were recorded. The mean (SD) successful intubation time with the Macintosh laryngoscope was significantly shorter compared with the McGrath laryngoscope, 50.0 (32.6) s vs 82.7 (80.0) s, respectively (p = 0.0003), despite the McGrath laryngoscope s having a lower intubation difficulty score and a superior glottic view. There were fi ve McGrath laryngoscope intubation failures, three owing to dif ficulty in passing the tracheal tube and two to equipment malfunction. Equipment malfunction is a major concern as a reliable intubating device is vital when faced with an airway crisis.